Deaerator Level Monitoring for Ketchup Processing Replacing Differential Level
We supplied a LN Potentiometric level sensor to the plant for their evaluation. The Plant had tried top down capacitance level probes in the past but had switched to differential level because they could not solve the erratic output cause by coating. While similar in form to capacitance probes, the LN technology emulates a slide potentiometer. Temperature changes do not impact the measurement and coatings from sticky or viscous product are ignored because they merely form a parallel connection with the otherwise very low-resistance stainless probe. What is measured is only the actual product between tube and vessel wall. For the customer, the stable accurate level measurement the LN provided allowed them to better control the deaerator level, eliminated discharge pump cavitation, improved timing pump flow control and reduced discharge pump maintenance. This application success lead to the ongoing retrofit of all their deaerators.